JC601 Digital Pressure Gauge
JC601 Digital Pressure Gauge adopts micropower components and unique software processing technology. Adopting Li-battery power, it could work 24 hours continuously for at least one year. With a 5-digit large crystal screen, the data could be shown more clearly.
- Price Term: FOB/CIF/EXW/DAP
- Package: Exported Wooden case or according to customer request
- Delivery Time: 7 to 10 working days
- Transports: By sea, by air, by high express
- Shipment Port: Shanghai Port
- Origin of Place: Jiangsu
- Warranty: 12 Months
Every JC 601 has passed the high and lower temperature aging thus ensuring more reliable precision. The product has been used in the industries of power, chemical, petrol, metallurgy, metering, etc.
- Low Power Consumption: 1pc disposable 3.6Vdc lithium battery
- 5-digit large LCD with backlight function
- 9 Pressure Units selectable
- Zero point, full-scale calibration function
- Adopting the special software treatment technology, it is unnecessary to preheat
- RS232 Communication Port can communicate with PC
- Temperature compensating ●Overload alarm function
- % indication with bar graph scale for visual reference
- Stainless steel bottom connection for robustness
- Equipped with rugged aluminum alloy cover
Apply pressure to the pressure transducer, it outputs an electrical signal. First, the corresponding electrical signal will be amplified. and then it will be converted to digital data by an A/D converter. Then the microprocessor sequentially collects data to modify and correct, the final result will be displayed on the LCD Display.
Please turn off the instrument before changing the battery. Open the cover, take out the battrery and change to a new one.
- Replace the battery if the battery icon appears.
- Remove the JC601 from the Ex-hazardous area before opening the cover .
- Use only the 3.6VDC Li battery.
- When replacing the battery, make sure the positive and negative are in the right direction.
2.The maintenance instrumnet
(1) Pay attention to the pressure range selected while measuring. Don’t allow it to work under an overpressure environment; Overpressure alarm limit of the gauge is 1.1 times of the full-scale;
(2) The using environment had to satisfied the manual demand, the user can operate manually.
(3) The gauge uses one piece of disposable DC3.6V lithium battery as its power supply, when the screen displays the battery symbol, it indicates that the gauge is lacks battery power, replace the battery promptly.
(4) Don’t use the gauge in HHE for a long time.