The accuracy of a turbine flow meter is an essential factor in many industrial and commercial operations. From monitoring water flow in industrial processes to measuring fuel consumption in automobiles, these meters’ accuracy can significantly impact the efficiency and effectiveness of operations. In this blog post, we will explore the various factors that can affect the accuracy of turbine flow meters.
What is Turbine Flow Meter?
Turbine flow meters are a type of velocity flow meter that has found widespread use in various industrial applications – including aerospace, cryogenic, and custody transfer – for high-accuracy measurements. The turbine flow meter is reliable for liquids and gases. The rotor’s diameter is slightly less than the inside diameter of the metering chamber, and its rotation speed is proportional to the volumetric flow rate. Turbine rotation can be detected by solid-state devices (reluctance, inductance, capacitive and Hall-effect pick-ups) or mechanical sensors (gear or magnetic drives).
How do Turbine Flow Meters work?
The measurement principle of any turbine flow meter depends on the mechanical exertion of a fluid upon a rotor/spindle, similar to how a fan turns when the wind blows on it. Turbine flow meters have specially manufactured blades that differ according to the application. An increase in blade rotation is proportional to a velocity increase. Since the surface area of the pipe is known, we can deduct the volumetric flow.
Applications of Turbine Flow Meters
- Water injection ·
- Two and Three Phase Separators ·
- Water wells ·
- Chemical injection systems ·
- Aerospace/ Military ·
- Engine Bed Testing ·
- Power Plants ·
- Building automation ·
- HVAC ·
- Cryogenics (extremely low temperatures) ·
- Tank Truck Unloading
What factors affect the accuracy of the Turbine Flow Meter?
The reasons behind the decrease in the accuracy of turbine flow meters during use are often the same. Specifically, these may include:
- Contamination of the medium due to particles, dirt, and other debris can have serious consequences. These impurities can cause wear and tear on bearings and shafts, leading to an increase in the gap between them, disrupting the dynamic balance of moving parts, reducing the rotating speed, and increasing the resistance of movement. Furthermore, when dirt enters the gap, the rotating speed reduces, and the displayed value of the instrument decreases, resulting in an error that can damage the fluid supplier. Therefore, to protect your system from these issues, it is crucial to ensure the medium remains free from contamination.
- Changes in the fluid temperature and pressure of the transmitter flow meter can lead to disastrous consequences, such as the liquid in the pipeline escaping the air it contains or the pressure in the pipeline is lower than the saturated vapor pressure of the fluid, resulting in part of the liquid becoming vapor. Additionally, outdoor gas can be sucked into the pipeline due to the low pressure of the medium, which flows with the measured liquid and causes a false positive reading on the meter, resulting in inaccurate results for the demand side of the fluid.
- The harmful errors caused by the fibrous or viscous impurities in the fluid attaching to the rotating part of the flow meter can result in significant inaccuracies in the indication value of the meter, which is unfavorable to the fluid supplier. This is due to the increased resistance to rotation caused by the impurities, which reduces the indication value. To prevent these errors, it is vital to remove any contaminants present in the fluid before the flow meter measures it.
- Under the standard turbine-type flow meter working principle, the working environment is often harsh, such as high temperature, vibration, electromagnetic field interference, dust, humidity, etc. , which can lead to malfunctions or failures of the turbine flow sensor and consequently cause positive or negative errors in the indication of the turbine flow meter. These errors may be subtle or even wholly invalid. To prevent such issues, it is essential to ensure the proper installation and maintenance of your turbine flow meter to ensure its accuracy and reliability.
For the above phenomenon, when it is more serious, it is easy to identify the issue and take the necessary corrective measures by comparing the functional status of the process. But unfortunately, in the early stages of the problem, it cannot be easy to detect without taking extraordinary measures. Therefore, it is important to identify such issues before they become too serious proactively.
The accuracy of turbine flow meters can be affected by various factors, such as the type of fluid, the pressure and temperature, the number of blades, the size of the turbine, and any foreign particles present in the fluid. Other factors that can influence the accuracy of turbine flow meters include the installation of the meter, the type of material used in the meter, the accuracy of the instruments used to measure the flow, type of flow meter, and the quality of the calibration process. Understanding these factors’ potential impacts can help ensure that the turbine flow meter performs accurately and reliably.
Contact BCST for inquiries.